SCAL2 Handheld Acoustic Emission Calibrator Series are designed and developed for the requirements of acquisition host calibration of acoustic emission detector, which is based on¡¶JJF1505-2015 Calibration Specification for Acoustic Emission Instrumentation¡·, and also can be used as a sample standard signal emission, in order to export standard acoustic emission signal waves, mainly used for calibration of the AE instruments. Scal2 series can export various standard parametric signals, can be used on calibrating AE indicators like amplitude¡¢rise time¡¢ring-down count and times of duration. Scale 2 series are applicable for calibration of AE instrument before use¡¢usability of AE sensors and AE preamplifier, and is especially applicable for distinguishing and testing the working condition of the AE systems quickly once on the testing environment site.
Displaying content on the screen
Battery powered and continuous working time is no less than 10 hours£»
(3) Connector type:BNC
(4) Working temperature: 0¡«40¡æ£»
(5) Storage temperature: 0¡«40¡æ£»
(6) Relative humidity: 30¡«90%£»
(7) Avoid external electromagnetic interference£»
£¨8£©Prevent vibration can meet GB/T 242310.2008 standard (or other equivalent standard).
£¨1£©Colorful screen, resolution ratio 320 x 240£»
£¨2£©Keyboard controls parameter setting and signal emission£»
£¨3£©Accuracy: range of amplitude ¡À0.3dB£¬frequency range ¡À0.5%
£¨1£©Settable variables £ºsignal waveform¡¢signal amplitude¡¢signal frequency, emission speed¡¢gain¡¢threshold;
£¨2£©Signal waveform£ºAE double exponential envelope¡¢AE triangular envelope¡¢impulse sine wave ¡¢continuous sine wave¡¢signal impulse wave£»
£¨3£©Signal amplitude of the sine wave£º30dB-100dB£¬
£¨4£©Signal amplitude of the AE waveform: 30-90dB£¬
£¨5£©Signal frequency of the continuous sine wave:10kHz¡«1MHz£¬0.1kHz£»
£¨6£©Signal frequency of AE waveform£º30kHz£¬60kHz£¬150kHz£¬300kHz
£¨7£©AE parametric indicators: rise time¡¢duration time£¬ring-down count£»
£¨8£©Emission rate: 1pps£¬10pps£¬100pps£¬1000pps£¬manual control;
£¨9£©Preamplifier gain: 0dB£¬26dB£¬40dB
£¨10£©After boot initialization, the system can automatically turn out the configuration which was set before closing.